|Place Of Origin:||Guangdong, China||Power:||Electronic|
|Temp Range:||RT+10°C ~70°C||Usage:||Universal Testing Machine|
|Light/ Lamp:||UV Lamp 40W||Humidity Range:||RH75%~95%|
uv testing equipment,
accelerated aging test chamber
UV accelerated weathering test chamber/uv weather test
1. test sample fixed to the test rack fixture. when the sample holder is not filled with the
sample faced with fluorescent light , you need to use the blackboard to fill the sample holder, keep
the chamber inner wall closed.
2 test temperature. Illumination can be 50,60,70 ℃ three temperatures, priority recommended
60 ℃; condensed phase temperature Degree is 50 ℃, the temperature uniformity tolerance of ± 3
3.Each condensation cycle selectable illumination light 4H, 4H condensation or 8H light, 4H
condensing: two kinds of cycles.
4 After the first light 400 ~ 450H, every row need to be replaced with a fluorescent lamp. Other
lamps in accordance with the light cycle and replace
Way to replace the lamp to compensate for losses caused by lamp aging. This mode of operation
is to ensure that each lamp life is between 1600 ~
5 When replacing the lamp, a water dish should be dry and clean, avoid the formation of scale.
UV Lamp Weathering Test Chamber.pdf
Contact Person: Nancy Shen